Principle Measurement by amperometric method with a diaphragm cell of 3 electrodes; wasted sample. The probe includes a CTN sensor for the temperature compensation.
Mounting / Recommendations The measuring at a constant flow rate requires the use of a specific cell (see data sheet 193-95). The complete assembly optimizes the operations.
Range: 0.001 to 20.0 g /L Operating pH: 1 to 7 Pressure limits: 1 bar as a maximum Temperature limits: 1 to 45 °C Flow rate limits: From 30 to 40 L /h (see the data sheet 193-95) Power supply: 12...30 V DC